Volumetric imaging of the fetal heart can be accomplished with a variety of commercially available ultrasonographic equipment. The modality is increasingly used in clinical practice, either to help solve difficult cases or to gain a different perspective on anomalies seen by conventional two-dimensional ultrasonography (2DUS). The objectives of this CME activity are:
Luis F. Goncalves, MD
Professor of Radiology and Obstetrics and Gynecology
Oakland University William Beaumont School of Medicine
Royal Oak, MI and Rochester MI
The Institute for Advanced Medical Education is accredited by the Accreditation Council for Continuing Medical Education (ACCME) to provide continuing medical education for physicians.
The Institute for Advanced Medical Education designates this enduring material for a maximum of 1 AMA PRA Category 1 Credit™. Physicians should only claim credit commensurate with the extent of their participation in the activity.
These credits are accepted by the American Registry for Diagnostic Medical Sonography (ARDMS).
For information on applicability and acceptance of continuing education credit for this activity, please consult your professional licensing board or other credentialing organization.
In order to complete this program you must have a computer with a recent browser version. You must also have the capability to display and print PDF files in order to view and print out your certificate. (Note: Your CME certificate is stored in your account and is available at any time.)
For any questions or problems concerning this program or for problems related to the printing of the certificate, please contact IAME at 802-824-4433 or email@example.com.
This activity is designed to be completed within the time designated. To successfully earn credit, participants must complete the activity during the valid credit period. To receive AMA PRA Category 1 Credit™, you must receive a minimum score of 70% on the post-test.
Follow these steps to earn CME credit:
Your CME credits will be archived in the account you create and can be accessed at any time.
In compliance with the Essentials and Standards of the ACCME, the author of this CME tutorial is required to disclose any significant financial or other relationships they may have with commercial interests.
Dr. Luis F. Goncalves discloses a relationship with GE Healthcare and with Philips Healthcare as a member of their speakers bureaus.
No one at IAME who had control over the planning or content of this activity has relationships with commercial interests.
- Three-dimensional (3D): term used to refer to volume datasets defined by 3 spatial dimensions (x, y, and z planes). Since only spatial information is acquired, these volume datasets are static by nature.
- Four-dimensional (4D): refers to volume datasets containing the 3 spatial dimensions plus the temporal dimension. These volume datasets can be displayed with motion and are extensively used for volumetric imaging of the fetal heart.
- Real-time: this term describes a negligible delay between acquisition and display.
- Rendering: technology that allows three-dimensional objects to be displayed on a two-dimensional (2D) screen.
- STIC: acronym for SpatioTemporal Image Correlation. This technology allows retrospective gating of the spatial volumetric data obtained from a beating fetal heart to the fetal heart rate. The process works by re-shuffling of the 2D frames acquired during a sweep through the fetal heart according to the phase of the cardiac cycle at which they were acquired. The result is a series of volume datasets, each representing a phase of the cardiac cycle, which are then played on the screen as a continuous cine loop.2
Volumetric images of the fetal heart can be obtained using mechanical 3-5 or matrix array transducers. 6-9
Mechanical volumetric transducers generally consist of a convex array mounted on a mechanical wobble. The wobble is electronically steered so that it can sweep through a region of interest in an automated fashion. As a result, a series of 2D images with precise x, y, and z spatial coordinates are obtained. Once the images are acquired, they are assembled into a volume dataset, which is then ready to be displayed on a 2D screen using one of several methods described in greater detail below.
Matrix array transducers are electronic transducers composed of thousands of elements arranged on a 2D matrix. All or a portion of the matrix elements can be fired simultaneously, producing a 3D pyramid of sound that can be visualized either as static 3D or as a dynamic real-time 4D volume. Alternatively, rows of elements can be fired in sequence, simulating the volume dataset acquisition process that occurs with mechanical probes.
Both mechanical volumetric and matrix array transducers are capable of producing dynamic volume datasets of the fetal heart using STIC technology.
Volume acquisition, regardless of transducer technology, begins with optimal 2D imaging. This step is of utmost importance when scanning the fetal heart. The goal is to have a final volume dataset with good spatial resolution, optimal contrast, and as high as possible temporal resolution. The following are tips to optimize the 2D image prior to volume acquisition:
- Select the transducer with the highest possible frequency for the scanning conditions, taking into account the position of the placenta and maternal body habitus. Obese patients with anterior placentas likely require the selection of a lower frequency transducer.
- Increase the contrast to emphasize a clear distinction between the endocardium, valvular apparatus, and blood pool. This is usually accomplished by adjusting the dynamic range and gain settings to the scanning conditions.
- Make the scanning sector as narrow as possible to maximize the frame rate. This step is particularly important if the volume datasets are acquired with color or power Doppler, since both will negatively impact the frame rate.
- After resolution, contrast, and frame rate are optimized, apply proper magnification.
Place the region of interest (ROI) box around the heart. The dimensions of the box, as seen on the screen, determine the amount of data to be acquired in the x- and y-planes. The key here is to pay attention to the width of the ROI. If the ROI is too wide, the frame rate drops, and the final volume dataset may suffer from low temporal resolution. Again, attention to detail in this step is even more important if the volume acquisition is performed with color or power Doppler. As a rule of thumb, select ROIs that include the heart, chest wall, and spine when acquiring volume datasets with gray scale only. If the acquisition involves color or power Doppler, narrow the ROI to include only the heart and great vessels.
The angle of acquisition determines the amount of information to be reconstructed in the z-plane. Acquisition angles vary depending on the manufacturer, and can generally be set between 5 and 45 degrees. A large acquisition angle provides more information; however, one needs to remember that fetuses may move during volume acquisition. The goal is to select the narrowest possible angle to include images from the superior mediastinum to the upper abdomen. The basic rules are “small angle for small fetus” and “large angle for large fetus”. It may take some trial and error before one feels comfortable in adjusting this parameter on the fly.
This parameter determines the number of frames that are acquired and incorporated into the final volume dataset. Ideally, longer acquisition time allows more frames to be acquired. This results in a higher spatial resolution, provided that the fetus does not move or moves little during acquisition. Therefore, if the fetus is in an ideal position and not moving, prefer a longer acquisition time. The reverse applies to fetuses that are actively moving. Depending on the manufacturer, acquisition speed ranges from 5 to 15 seconds.
Once a high quality volume dataset is acquired, it is time to visualize it on the screen and to able to manipulate it effectively to extract as much diagnostic information as possible. In the following sections, the most commonly used visualization methods are reviewed.
Multiplanar display is the most commonly used method to visualize volumetric images, regardless of the imaging modality used for acquisition (e.g., computerized tomography, magnetic resonance imaging, ultrasonography, etc.). The screen is usually split into three panels, representing the original plane of acquisition plus two additional reconstructed planes (Figure 1).
A reference dot or cross is usually used to indicate the point of intersection between the three planes. This reference dot can be moved on the screen with the aid of a mouse, allowing the same structure to be simultaneously visualized in three orthogonal planes. This technology makes possible virtual navigation through a volume dataset (Figure 2).
A few techniques have been previously described that may aid the examiner in performing a more purposeful navigation of the volume dataset. One such technique has been called “spin technique”10 It consists in rotating the volume dataset around the y- or x-axes using the reference dot as a pivot point. The structure being interrogated “opens up” and nicely displays the relationships with other surrounding structures. Another commonly used technique, which was specifically developed to simultaneously display the left and right outflow tracts is known as the “three-step technique”.4, 11 It consists of: 1) re-orienting the fetus on the screen such as that the spine is down; 2) anchoring the reference dot in the middle of the ventricular septum (step 1, Figure 3A); 3) rotating the volume dataset around the y-axis until the continuity between the anterior wall of the aorta and the ventricular septum are seen (step 2, Figure 3B); and 4) moving the reference dot to the level of the aortic valve so that the short axis view of the right ventricular outflow tract is seen simultaneously on the sagittal orthogonal plane (step 3, Figure 3C).
This method automatically slices the volume dataset in any plane chosen by the examiner.12, 13 Slices are displayed in sequence on the screen. One of the frames (usually the first frame or a side frame) is used as a “scout view” depicting the exact position of the slices within the volume dataset (Figure 4).
Rendering is a technology that permits realistic display of 3D volume datasets on a 2D screen (Figures 5 and 6). This is the same technology used to display 3D video games on a flat screen. The examiner can control how the image is displayed and can choose seeing just the surface of the object (surface mode), display the brightest echoes (maximum intensity projection), display the darkest echoes (minimum intensity projection), or display the average intensity of the echoes along the projection path (x-ray mode).
Rendering using color Doppler, Power Doppler, Inversion Mode and B-Flow Imaging
Any of these modalities can be used to provide “virtual contrast” to the cardiac chambers and blood vessels, so that these structures can be displayed like digital “casts”.14-22 Figures 7, 8, and 9 provide examples of rendered images of normal fetal hearts using each of these technologies.
The following are clinical examples that illustrate the capabilities of volumetric imaging of the fetal heart.
A few studies have addressed the feasibility or applicability of volumetric imaging of the fetal heart in clinical practice. Uittenbogaard et al.23 evaluated volumetric images of the fetal heart obtained from 148 patients seen in the clinical setting. All fetuses were high risk and were examined by 2 experienced examiners. No more than 4 attempts at obtaining volume datasets were allowed during each exam. Successful acquisition was possible in 76% of the cases. Twenty-five percent of those were considered high quality, 40% of sufficient diagnostic quality, and 35% were non-diagnostic. Factors associated with high quality volume datasets included a lower body mass index (23.8 kg/m2 vs. 26.5 kg/m2, p=0.04) and posterior placentas (56.0% vs. 30.3%, p=0.05). Visualization rate for cardiac structures was higher for high quality datasets when compared to those of only sufficient diagnostic quality. In a similar study, Cohen et al.24 determined how frequently satisfactory images for fetal screening could be obtained in nonobese patients scanned at 18 to 22 weeks of gestation, within a maximum examination allotted time of 45 minutes. Satisfactory images of the 4-chamber view, left ventricular outflow tract (LVOT) and right ventricular outflow tract (RVOT) were obtained from the volume datasets by two experienced examiners in 91 to 96.4%, 77.5 to 85.6%, and 80.2 to 87.4% of the time, respectively. A lower frequency of satisfactory images was noted for fetuses with the spine anteriorly positioned, and also when the placenta was anterior.
Volumetric imaging appeals to those interested in telemedicine applications.25-28, 5 The basic concept is that, particularly in areas with a lack of expertise in fetal echocardiography, volume datasets of the fetal heart could be acquired at the point of care by ultrasound services providers and transmitted to centers with expertise for either a second opinion or definitive diagnosis. Viñals et al.28 provided instructions to 2 remote examiners by email on how to properly acquire STIC volume datasets of the fetal heart, which were subsequently uploaded to a web server. The volumes were reviewed by a sonologist with experience in fetal echocardiography. All 47 normal fetuses were correctly identified as normal and the 3 fetuses with cardiac defects [ventricular septal defect (VSD), D-transposition of the great arteries (D-TGA), and atrioventricular septal defect (AVSD)] were correctly detected.
More recently, a multicenter international study evaluated the accuracy of remote interpretation of volume datasets collected at several institutions and interpreted by experts who were blinded for the exam indications and outcome. Ninety volume datasets of normal fetuses and fetuses with congenital heart disease were reviewed by experts at 7 different institutions. The sensitivity and specificity for diagnosis of congenital heart disease (CHD) were 93% and 96%, with excellent interobserver agreement (Cohen kappa = 0.97).25
A direct comparison between 2D and volumetric fetal echocardiography has also been performed. Bennasar et al.29 obtained volumes of the fetal heart during 2D fetal echocardiography in 342 fetuses with suspected CHD. Volume datasets were analyzed in a blinded fashion 1 year after the exam. There was no difference in the overall accuracy of 2D fetal echocardiography when compared to volumetric imaging (94.2% vs. 91%, p > 0.05). There were 9 false-negative diagnoses with volumetric imaging: 8 cases of VSD and 1 case of aortic arch interruption. Two-dimensional fetal echocardiography did not detect 2 VSDs and 1 case of persistent left superior vena cava. False positive diagnoses for volumetric imaging included: 10 cases of VSDs, 4 cases of coarctation of the aorta, 2 cases of persistent left superior vena cava, 1 case of pulmonic stenosis, and 1 case of rhabdomyoma. False-positive diagnoses by 2D fetal echocardiography were: 1 case of VSD, 4 cases of coarctation of the aorta, 1 case of tricuspid dysplasia, and 1 case of ostium primum atrial septal defect. The authors concluded that, in a high-risk population, volumetric imaging of the fetal heart is as accurate as 2D fetal echocardiography.
Most, if not all, of the pitfalls which occur in volumetric imaging of the fetal heart result from problems during acquisition. Motion and unfavorable fetal position are the most common obstacles to high quality volume datasets. Fetal motion, breathing and hiccups are difficult to control and are associated with artifacts that are best seen on the B-panel of a multiplanar display.
Obtaining good quality volume datasets requires being alert to good opportunities. These present when the fetus is in ideal position (with the spine down), no limbs in front of the chest, and holding still. Sometimes, high quality volume datasets can be obtained from a lateral approach. If the spine is up, excessive shadowing prevents high quality volume datasets to be acquired. Again, there is no way to control the fetus and the best approach is to be prepared when the opportunity presents. This is accomplished by performing the entire examination using volumetric probes. Once the fetus is not moving and the acoustic window to the fetal heart is ideal, the examiner should acquire as many volumes as possible in sequence. These can always be reviewed later and chances are that at least one or two of them will be either diagnostic or of high quality.
Maternal abdominal wall motion due to breathing can be another source of artifact, often times difficult to control. In this situation, we commonly ask the mothers to take shallower breaths during acquisition or to stop breathing momentarily.
Volumetric imaging of the fetal heart is now moving from the highly specialized research laboratories into mainstream clinical practice. This article reviewed the fundamentals of the technology, including volume dataset acquisition and display. Illustrative examples of common congenital cardiac disease were presented to illustrate the potential of the technology. Studies pertaining to the applicability of volumetric imaging of the fetal heart to clinical practice were reviewed, collectively showing that experts can diagnose congenital heart disease with a high degree of accuracy by relying on the volume datasets, suggesting that the technology can be used not only as a problem solving tool, but also as a means to provide access to expert examiners through telemedicine applications
1. Deng J. Terminology of three-dimensional and four-dimensional ultrasound imaging of the fetal heart and other moving body parts. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2003;22:336-44.
2. Nelson TR, Pretorius DH, Sklansky M, Hagen-Ansert S. Three-dimensional echocardiographic evaluation of fetal heart anatomy and function: acquisition, analysis, and display. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 1996;15:1-9 quiz 11-2.
3. DeVore GR, Falkensammer P, Sklansky MS, Platt LD. Spatio-temporal image correlation (STIC): new technology for evaluation of the fetal heart. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2003;22:380-7.
4. Goncalves LF, Lee W, Chaiworapongsa T, et al. Four-dimensional ultrasonography of the fetal heart with spatiotemporal image correlation. American journal of obstetrics and gynecology 2003;189:1792-802.
5. Vinals F, Poblete P, Giuliano A. Spatio-temporal image correlation (STIC): a new tool for the prenatal screening of congenital heart defects. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2003;22:388-94.
6. Goncalves LF, Espinoza J, Kusanovic JP, et al. Applications of 2-dimensional matrix array for 3- and 4-dimensional examination of the fetus: a pictorial essay. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2006;25:745-55.
7. Maulik D, Nanda NC, Singh V, et al. Live three-dimensional echocardiography of the human fetus. Echocardiography 2003;20:715-21.
8. Sklansky MS, DeVore GR, Wong PC. Real-time 3-dimensional fetal echocardiography with an instantaneous volume-rendered display: early description and pictorial essay. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2004;23:283-9.
9. Sklansky MS, Nelson T, Strachan M, Pretorius D. Real-time three-dimensional fetal echocardiography: initial feasibility study. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 1999;18:745-52.
10. DeVore GR, Polanco B, Sklansky MS, Platt LD. The 'spin' technique: a new method for examination of the fetal outflow tracts using three-dimensional ultrasound. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2004;24:72-82.
11. Rizzo G, Capponi A, Muscatello A, Cavicchioni O, Vendola M, Arduini D. Examination of the fetal heart by four-dimensional ultrasound with spatiotemporal image correlation during routine second-trimester examination: the 'three-steps technique'. Fetal diagnosis and therapy 2008;24:126-31.
12. Devore GR, Polanko B. Tomographic ultrasound imaging of the fetal heart: a new technique for identifying normal and abnormal cardiac anatomy. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2005;24:1685-96.
13. Goncalves LF, Espinoza J, Romero R, et al. Four-dimensional ultrasonography of the fetal heart using a novel Tomographic Ultrasound Imaging display. Journal of perinatal medicine 2006;34:39-55.
14. Chaoui R, Hoffmann J, Heling KS. Three-dimensional (3D) and 4D color Doppler fetal echocardiography using spatio-temporal image correlation (STIC). Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2004;23:535-45.
15. Espinoza J, Romero R, Kusanovic JP, et al. Prenatal diagnosis of coarctation of the aorta with the multiplanar display and B-flow imaging using 4-dimensional sonography. J Ultrasound Med 2009;28:1375-8.
16. Goncalves LF, Espinoza J, Lee W, et al. A new approach to fetal echocardiography: digital casts of the fetal cardiac chambers and great vessels for detection of congenital heart disease. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2005;24:415-24.
17. Goncalves LF, Romero R, Espinoza J, et al. Four-dimensional ultrasonography of the fetal heart using color Doppler spatiotemporal image correlation. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2004;23:473-81.
18. Hata T, Dai SY, Inubashiri E, et al. Four-dimensional sonography with B-flow imaging and spatiotemporal image correlation for visualization of the fetal heart. Journal of clinical ultrasound : JCU 2008;36:204-7.
19. Lee W, Espinoza J, Cutler N, Bronsteen RA, Yeo L, Romero R. The 'starfish' sign: a novel sonographic finding with B-flow imaging and spatiotemporal image correlation in a fetus with total anomalous pulmonary venous return. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2010;35:124-5.
20. Volpe P, Campobasso G, De Robertis V, et al. Two- and four-dimensional echocardiography with B-flow imaging and spatiotemporal image correlation in prenatal diagnosis of isolated total anomalous pulmonary venous connection. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2007;30:830-7.
21. Volpe P, Campobasso G, Stanziano A, et al. Novel application of 4D sonography with B-flow imaging and spatio-temporal image correlation (STIC) in the assessment of the anatomy of pulmonary arteries in fetuses with pulmonary atresia and ventricular septal defect. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2006;28:40-6.
22. Volpe P, Tuo G, De Robertis V, et al. Fetal interrupted aortic arch: 2D-4D echocardiography, associations and outcome. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2010;35:302-9.
23. Uittenbogaard LB, Haak MC, Spreeuwenberg MD, Van Vugt JM. A systematic analysis of the feasibility of four-dimensional ultrasound imaging using spatiotemporal image correlation in routine fetal echocardiography. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2008;31:625-32.
24. Cohen L, Mangers K, Grobman WA, Platt LD. Satisfactory visualization rates of standard cardiac views at 18 to 22 weeks' gestation using spatiotemporal image correlation. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2009;28:1645-50.
25. Espinoza J, Lee W, Comstock C, et al. Collaborative study on 4-dimensional echocardiography for the diagnosis of fetal heart defects: the COFEHD study. Journal of ultrasound in medicine : official journal of the American Institute of Ultrasound in Medicine 2010;29:1573-80.
26. Viñals F. Current experience and prospect of internet consultation in fetal cardiac ultrasound. Fetal diagnosis and therapy 2011;30:83-7.
27. Viñals F, Ascenzo R, Naveas R, Huggon I, Giuliano A. Fetal echocardiography at 11 + 0 to 13 + 6 weeks using four-dimensional spatiotemporal image correlation telemedicine via an Internet link: a pilot study. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2008;31:633-8.
28. Viñals F, Mandujano L, Vargas G, Giuliano A. Prenatal diagnosis of congenital heart disease using four-dimensional spatio-temporal image correlation (STIC) telemedicine via an Internet link: a pilot study. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2005;25:25-31.
29. Bennasar M, Martinez JM, Gomez O, et al. Accuracy of four-dimensional spatiotemporal image correlation echocardiography in the prenatal diagnosis of congenital heart defects. Ultrasound in obstetrics & gynecology : the official journal of the International Society of Ultrasound in Obstetrics and Gynecology 2010;36:458-64.